Integra® Meshed Bilayer Wound Matrix
Integra® Meshed Bilayer Wound Matrix is an advanced bilayer matrix device consisting of a porous, three-dimensional dermal layer and a temporary epidermal layer. The dermal layer (collagen and glycosaminoglycan) is designed with a controlled porosity and defined degradation rate that provides a scaffold for cellular invasion and capillary growth. The epidermal layer (polysiloxane) is pre-meshed.
• Saves OR time from meshing
• Designed to be used with negative pressure wound therapy
• Meshed design allows better conformity for various anatomical sites
• Provides immediate wound closure
• Room temperature storage
Integra® Meshed Bilayer Wound Matrix is indicated for the management of wounds including: partial- and full-thickness wounds, pressure ulcers, venous ulcers, diabetic ulcers, chronic vascular ulcers, surgical wounds (donor sites/grafts, post-Mohs surgery, post-laser surgery, podiatric, wound dehiscence), trauma wounds (abrasions, lacerations, second-degree burns, and skin tears) and draining wounds. Integra® Meshed Bilayer Wound Matrix may be used in conjunction with negative pressure wound therapy. The device is intended for one-time use.
This device should not be used in patients with known sensitivity to bovine collagen or chondroitin materials. This device is not indicated for use in third-degree burns.
Do not resterilize. Discard all opened and unused portions of the device. Do not use if package seal is broken.
Complications are possible with the use of wound dressings including infection, chronic inflammation, allergic reaction, excessive redness, pain or swelling.
Room temperature with a 24-month shelf life.
|Indicated for chronic vascular ulcers||●|
|Indicated for diabetic ulcers||●|
|Indicated for draining wounds||●|
|Indicated for partial- and full-thickness wounds||●|
|Indicated for pressure ulcers||●|
|Indicated for surgical wounds||●|
|Indicated for third-degree burns|
|Indicated for trauma wounds||●|
|Indicated for venous ulcers||●|
|Variety of sizes||●|
|Educational Material Available||●|
|Free Samples/Trials Available||●|
|Published Clinical Article Available|
Using aseptic technique, peel open the outer pouch and gently drop the inner foil pouch onto a sterile field or surface. Using sterile technique, place foil pouch flat and peel open the inner pouch. Remove product, including the protective polyethylene sheets. While holding the product with the tab, remove one polyethylene cover sheet. Turn the product and remove the second polyethylene cover sheet. Using the tab, the product can now be placed into a basin containing a sterile saline solution. Carefully remove the tab from the product while rinsing for 1-2 minutes. Keep product in the basin until application.
Prepare wound bed using standard methods to ensure wound is free of debris and necrotic tissue. If necessary, surgically debride the wound to ensure the wound edges contain viable tissue. Cut the device to size and apply immediately following wound bed preparation. Note: It is critical that the collagen layer be in direct contact with the prepared wound. The silicone layer, identified by the black threads, must be placed out (away from the wound bed). Do not apply upside down; the black threads must be clearly visible.
Integra® Meshed Bilayer Wound Matrix should be firmly secured using surgical staples, sutures, or other mechanical means. Any air bubbles should be carefully removed to maintain contact with the wound. After application, use appropriate secondary dressings to maintain device adherence and protect the wound area. The optimum secondary dressing is determined by wound location, size, depth and user preference.
Removal of silicone layer
If edges of Integra® Bilayer Wound Matrix are loose before full healing has occurred, the silicone can be trimmed away from the loose areas until the entire wound has healed. Remove the silicone layer of the Integra® Meshed Bilayer Wound Matrix when the tissue underneath is healed, typically 14 to 28 days.
The Matrix may be loose in spots. Remove by starting at one corner and pull gently. The silicone layer will peel off healed tissue relatively easily. Caution: If bleeding occurs, or if patient complains of excessive pain, stop and wait 1 to 2 additional days. Forced removal may result in wound re-injury.
Change the secondary dressings as needed. Frequency of secondary dressing change will be dependent upon volume of exudate produced, type of dressing used and the clinician's need to inspect the wound bed for signs of infection or healing.
Integra LifeSciences, a worldwide leader in regenerative medicine, is dedicated to improving the quality of life for patients through the development of cost-effective solutions used to treat millions of patients every year. With over two decades of clinical efficacy and a reputation for delivering quality, Integra's advanced wound care products are a premiere solution for clinicians.