REZair™ skin and wound aeration devices connect to all existing respiratory sources and continuously distribute air/oxygen to the patient's wound, unpressurized. Unique non-occludable technology operates even under extreme patient weights and fecal incontinence.
• Only wound aeration device on the market that connects to, and operates with, standard respiratory oxygen/air sources at respiratory high and low flow rates
• Easy to use, takes a few seconds to apply and provides continuous therapy
• Treatment can be started in clinical setting and continued at home or other long-term setting even when the clinician is not present
Do not adhere device to skin.
The device is packaged in individual pouches and can be stored in a cool, dry environment.
|Contact manufacturer for more details||●|
|Educational Material Available||●|
|Free Samples/Trials Available||●|
|Published Clinical Article Available|
Moderate/Highly Exudating Wounds
Clean wound fully. Do not apply topicals. Verify O2 connections are fully sealed, insert and adhere device into diaper, or apply around leg wound and secure with wrap bandage. Set air/oxygen source to clinician-preferred flow settings.
Recommended flow Settings: Pressure: 2-4 psi. Volume: 2-7 liters.
NOTE: Device may be operated safely, without damage, up to 10 psi. Connect REZair™ device to oxygen tubing and turn on respiratory oxygen.
Change whenever dressing or diaper is being changed.
REZAir™ can be used with any wound dressing or diaper.
Quality Specification: ISO-13485/ ISO-9001.
Sterility: Clean, Non-sterile. Produced in class 7 cleanrooms to FDA quality standards.